Heart attack? Wait longer for an ambulance: Huge 999 shake-up means slower response times for millions * Major overhaul of the ...
If you register when you purchase it,
C-Beam Developments will give you an unconditional 2 year replacement!
If your ASPOD is damaged in anyway, just pop it in the post and we will replace it FREE OF CHARGE
Opening Your Aspod®
PLEASE TAKE CARE WHEN OPENING YOUR ASPOD, EXCESSIVE FORCE COULD DAMAGE THE HINGE!
1: Hold eyelet between thumb and forefinger
2: Put index finger of the other hand under one of the protruding tabs on the outer edge of the Aspod lid
3: GENTLY AND SLOWLY pull lid upwards.
PLEASE LEAVE FOAM IN PLACE
Closing Your Aspod®
1: Bring front and rear together. Apply even pressure to areas around protruding tabs
2: Apply pressure just below eyelet until another click is heard
If no click is heard, hold both halves together placing thumb below eyelet, and push the eyelet forward towards you.
Or try this method :
Load the Aspod® with aspirin and hold it in front of you, with the Aspod® logo facing. You will see the eyelet in the centre with two ‘ears’, one on either side of the eyelet.
With your left thumb in front and index finger behind the left hand ‘ear’ apply pressure until a click is hear/or felt. Repeat this with the right hand ‘ear’.
Then with index finger behind where the eyelet joins the case apply pressure to the front top of the lid with the thumb and the final click will be heard/felt – The Aspod is now locked correctly.
Due to the minute tolerances involved in precision moulding, the final locking click on some ASPOD's may be 'felt' rather than heard. This does not indicate a faulty product.